宫腔深度 5.5- 6.0 cm 6.07.0.cm 7.0- 7.5 cm 7.5 cm以上 [Product Specifications] 5.5- 6.0 cm 6.07.0.cm 7.0- 7.5 cm 7.5 cm or more
格 GCu 200-18x20 GCu 200--20 x22 GCu 200-22X24 GCu 200- -22x24 Product specifications GCu 200-18x20 GCu 200--20 x22 GCu 200-22X24 GCu 200- -22x24
宫腔形宫内节育器由06Cr19Ni10不锈钢丝和纯度为99%的纯铜线组成。 [Product structure] Uterine intrauterine device is composed of 06Cr19Ni10 stainless steel wire and 99% pure copper wire. The copper surface area is nominally 200% mm2. Disposable products, sterilized with ethylene oxide. 腔 形宫内节育器根据妇女子宫腔的生理解剖形状及生理变化动态等特点而研发制造，该产品采用不锈钢螺旋管内灌装铜螺旋管构成近似于子宫腔的形态，放置于妇女子宫腔内作避孕用。 The intrauterine device is developed and manufactured according to the characteristics of the anatomical shape of the woman's uterine cavity and the dynamics of physiological changes. The product uses a stainless steel spiral tube filled with copper spiral tubes to form a shape similar to the uterine cavity and is placed in the woman's uterine cavity. For contraception.
产品性能 】 [ Product performance ]
宫 腔收缩时形态变化，具有防止脱落的作用，亦不易嵌顿，安全**、放置方便 ;其避孕原理是:通过铜离子在子宫内持续释放，增加了对精子的毒性，改变了 宫 腔内的环境，能抵制精子穿透宫颈粘液和延缓卵细胞通过输卵管，并能阻止孕卵在内膜的附着和植入，使受精卵不能着床而达到避孕的目的。 The uterine angiography of the intrauterine device is adapted to the morphological changes of most uterine cavity during contraction. It has the effect of preventing shedding, and is not easy to incarcerate. It is safe and convenient to place . Its contraceptive principle is: copper ion in the uterus Continuous release inside the body increases the toxicity to the sperm, changes the environment within the uterine cavity , resists sperm from penetrating cervical mucus and delays the passage of egg cells through the fallopian tubes, and prevents the attachment and implantation of the intimal layer of pregnant eggs, making fertilized eggs unable Implanted to achieve the purpose of contraception.
适用范围 】 宫腔形宫内节育器放置于妇女子宫腔内作避孕用。 [ Scope of application ] The intrauterine device is placed in a woman's uterine cavity for contraception.
禁忌症 】 同宫内节育器手术常规。 [ Contraindications ] Same as IUD surgery.
； 药物流产后月经还未潮者 ； 宫颈癌患者 ； 盆腔炎患者 ； 化脓性宫颈炎患者 ； 重度宫颈糜烂患者 ； 子宫脱垂者 ； 双子宫、子宫纵隔、双角子宫者。 Prohibited objects: after infectious abortion: vaginal bleeding of unknown cause ; those who have not had menstruation after medical abortion ; patients with cervical cancer ; patients with pelvic inflammatory disease ; patients with purulent cervicitis ; patients with severe cervical erosion ; uterine prolapse ; Uterine mediastinum, two-cornered uterus.
放置时间 】 月经期第 3天起*月经干净后3 - 7天。 [ Leaving time ] From the 3rd day of menstrual period * 3-7 days after menstruation is clean . Immediately after abortion and suction curettage and forceps.
使用年限 】 宫腔形宫内节育器使用年限为5-8年。 [ Service life ] The service life of intrauterine device is 5-8 years.
放置方法 】 [ Placement method ]
①. Before the placement of the woman, determine the position of the uterus, use cervical forceps to hold the cervix and pull it slightly to make the uterus flat, and then use the uterine probe to measure the longitudinal diameter of the uterine cavity.
②. According to the length of the uterine cavity, select a suitable intrauterine device of the intrauterine shape, adjust the curvature of the placement device and the position of the positioning block so that it corresponds to the measured length of the uterine cavity.
③ Hold the placement tube firmly with your left hand, and slowly and continuously pull the fixing wire under the placement tube down with your right hand, and close the upper arm of the intrauterine intrauterine device inward to pull it into the placement tube.
④. Hold the placement tube firmly with your left hand, and gently push the putter pusher forward with your right hand to make the top of the uterine intrauterine device's arc protrude from the placement tube by 1 mm, and pull out the fixing wire.
⑤Pull the cervical forceps with your left hand, and place the device with the IUD into the bottom of the uterine cavity in your right hand to keep the device in close contact with the bottom of the uterus; fix the pusher of the device and move it gently outwards Place the tube (time greater than 10 seconds), make the IUD in the placed tube pop out of the placed tube, and place it on the bottom of the uterine cavity.
⑥. After slowly withdrawing the pusher of the placer, slowly withdraw the placement tube, and the placement is completed.
⑦. One month after the operation, a B-ultrasonography was performed after menstrual cramps to determine whether the intrauterine intrauterine device was placed at the bottom of the uterine cavity to meet the contraceptive requirements.
宫腔形宫内节育器应由受过专业放置技术培训的医生放置 ; 1.Intrauterine IUDs should be placed by doctors trained in professional placement techniques ;
子宫腔置 于 宫腔形宫内节育器后两周内禁止性生活和盆浴，保持外阴清洁 。 2. The uterine cavity is forbidden to have sex and pelvic bath for two weeks after being placed in the uterine-shaped intrauterine device to keep the vulva clean .
子 宫 腔置于宫腔形宫内节育器后，可能出现在腹不适应或轻微出血，数天后症状会 自 然消失或好转 : 3. After the uterine cavity is placed in the intrauterine device, it may appear that the abdomen is unsuitable or slightly bleeding, and the symptoms will naturally disappear or improve after a few days :
周内不能做重体力劳动及剧烈运动 : 4, can not do heavy physical work and strenuous exercise within a week :
子宫腔置于宫腔形宫内节育器后应定期复查，可利用 B超或X光检 查 宫腔形 宫 内 节 育器的位置。 5. After the uterine cavity is placed in the uterine cavity IUD, it should be reviewed regularly. The position of the uterine cavity IUD can be checked by B-mode or X-ray .
有频繁脱环史或带器妊娠的妇女、哺乳期妇女在医生指导下慎用。 6, women with a history of frequent out-of-loop or pregnancy with lactation, lactating women with caution under the guidance of a doctor.
产品灭菌 】 [ Product sterilization ]
After being sealed and packaged, the intrauterine device is sterilized with ethylene oxide gas, and the sterilization period is three years.
(或封口纸)封口处。 The production date, sterilization batch number and expiration date can be found on the sealing place of the packaging bag (or sealing paper).
Jinan Kanghui Trading Company
Jinan Kanghui Trading Company